Vitreoretinal instruments: vitrectomy cutters, endoillumination and wide-angle viewing systems
© The Author(s) 2016
Received: 30 May 2016
Accepted: 28 October 2016
Published: 5 December 2016
There have been many advances in vitreoretinal surgery since Machemer introduced the concept of pars plana vitrectomy, in 1971. Of particular interest are the changes in the vitrectomy cutters, their fluidics interaction, the wide-angle viewing systems and the evolution of endoillumination through the past decade and notably in the last few years. The indications of 27-gauge surgery have expanded, including more complex cases. Cut rates of up to 16,000 cuts per minute are already available. New probe designs and pump technology have allowed duty cycle performances of near 100% and improved flow control. The smaller vitrectomy diameter can be positioned between narrow spaces, allowing membrane dissection and serving as a multifunctional instrument. Enhanced endoillumination safety can be achieved by changing the light source, adding light filters, increasing the working distance and understanding the potential interactions between light and vital dyes commonly used to stain the retina. Wide-angle viewing systems (contact, non-contact or a combination of both) provide a panoramic view of the retina. Non-contact systems are assistant-independent, while contact systems may be associated with better image resolution. This review will cover some current aspects on vitrectomy procedures, mainly assessing vitrectomy cutters, as well as the importance of endoillumination and the use of wide-angle viewing systems.
KeywordsEndoillumination Vitrectomy cutters Wide-angle viewing systems
Over 45 years ago, the idea of removing vitreous through a smaller aperture, with minimum trauma to the anterior compartment, was inspiring and revolutionary. In 1971, Machemer et al. introduced the concept of pars plana vitrectomy. The first vitreous cutter consisted of a micromotor which activated a drill bit inside a hypodermic needle, adapted on a plastic syringe and powered by a regular battery [1, 2]. The next significant step was taken by Conor O’Malley and Ralph Hein, which developed the three-port vitrectomy with a 20-gauge system as well as a lightweight, reusable, bellows-driven, pneumatic, axial cutter driven by the Ocutome 800 console (Berkley Bioengineering, 1972) . Since those humble beginnings there have been innumerable advances in vitrectomy surgery. Significant improvements have occurred not only with vitrectomy probes, which go faster and are smaller, but also in countless other aspects of our surgical environment including fluidics, endoillumination, handheld instrumentation, wound construction, console design and viewing systems, among others. As technology continues to improve it is important for surgeons to understand the implications that each of these innovations will have on their surgical performance and also how these variables will interact with each other. This article will review some advances in vitrectomy technology focusing on vitreous cutters, endoillumination and wide-angle viewing systems.
Since the 1970s, vitreous cutters have been modified to achieve high performance surgeries while maintaining safety. Numerous components such as the cutter size, cutting speed, port geometry, blade design and duty cycle, can alter the surgical efficiency and impact the postoperative results.
One of the major advances in retinal surgery is the reduced size of cutters. Smaller vitrectomy probes allowed the transition to the microincision vitrectomy system (MIVS), introduced in 2002 by Fujii et al., using 25-gauge instruments and followed by Eckardt in 2005, with 23-gauge cutters [4–6]. Subsequent years have shown that the modern MIVS have numerous advantages over 20-gauge vitrectomy and are associated with better patient comfort, less conjunctival scarring, less postoperative inflammation and earlier visual recovery [7–11]. More recently, 27-gauge instruments released by different companies have shown encouraging results [7–9, 12, 13].
In 2010, the first series of cases using 27-gauge vitrectomy probes, with selected macular diseases and non-complicated vitreous hemorrhages, was published by Oshima et al. . Although the fluid dynamics and cutting efficiency of the 27-gauge probes were lower when compared to the 25-gauge system, important aspects with the new tested devices were no need to convert to larger gauge instruments during the procedure and achievement of self-sealing sclerotomies, with no postoperative hypotony or endophthalmitis, which were observed in some of the initial studies using 23-gauge and 25-gauge probes [8, 14–17].
Nowadays, the indications for 27-gauge surgery have expanded. More complex cases including diabetic retinopathy, rhegmatogenous retinal detachment and nucleus fragments removal have been managed with the next generation of 27-gauge cutter probes [7, 9, 12–14, 18]. The thinner instrument can be safely introduced into smaller spaces between membranes and retina, serving as a multifunctional device and facilitating tissue dissection [7, 14, 18]. The decreased stiffness is still noted when compared to other gauges (i.e., half of the 25+ gauge stiffness), but the rigidity has been improved by shorter length needles or the introduction of a stiffness sleeve, such as the one released by Alcon . Khan et al. published the largest case series using the 27-gauge vitrectomy system in 2016. This multicenter study enrolled 95 eyes that underwent 27-gauge vitrectomy surgery, using the Constellation Vitrectomy 27+ Total Plus Pak (Alcon, TX, USA) probe for a variety of conditions. No intraoperative complications and no conversions to larger gauge instrumentation were required. Overall, the visual acuity improved from the baseline and the surgeons mentioned no complaints on instrument rigidity.
A substantial increase in cutter speeds has occurred since Machemer introduced the vitreous infusion suction cutter (VISC), in the 1970s. Current vitreous cutters are capable of delivering cut rates of up to 16,000 cpm (cuts per minute), depending on the vitrectomy platform (e.g., EVA; DORC International) [21, 22]. Although other components are involved in the vitreous flow rate (e.g., duty cycle), faster cutting speeds are generally associated with increased vitreous removal and therefore, surgery efficiency and shorter procedure time [20, 23]. The vitreous has an unpredictable flow behavior, difficult to characterize, due to its semi solid structure, composed of water, collagen fiber and hyaluronic acid  and differently from the balanced salt solution (BSS), that is easily aspirated, the vitreous requires cutting before going through the probe . Therefore, a high cutting rate is desirable. The chopped vitreous has a lower viscosity than intact gel and is more easily aspirated even in reduced diameter instruments . In addition, for the same flow rate, the higher the cut rate, the smaller the amount of vitreous (“bite size”) aspirated into the cutter, reducing both vitreous and retinal traction [19, 26]. Teixeira et al, assessing different 20, 23 and 25-gauge vitrectomy probes under porcine vitreous were able to demonstrate a decrease in retinal traction for every 500 cpm increase in the cut speed . Rizzo et al. have also shown a lower rate of iatrogenical retinal breaks when performing surgery with a high speed vitrectomy system (5000 cpm) when compared to a lower cut speed machine . Recently, Pavlidis et al., assessing a two-dimensional cutter probe (see next section), capable of cutting rates equivalent to 16,000 cpm in the DORC EVA (DORC International) platform, suggested that the higher cutting speed helped to ensure a faster vitreous removal when the performance was compared to a standard single port cutter of the same gauge . The liquefaction and excision of the vitreous body using an ultrasound, is another concept under investigation, with a prototype developed by Bausch and Lomb (Baush and Lomb, St. Louis, MO, USA). Initial reports under porcine eyes have shown promising results using 23 and 25-gauge probes, with no macroscopic retinal defects and no compromise of the BSS and vitreous flow rate when using different ultrasound powers, regardless of the gauge size [22, 28, 29].
Cutter port and blade design
The vitreous cutter port geometry and blade design may have a great influence in vitrectomy surgery from the fluidics and safety standpoints . DeBoer et al. demonstrated that increasing the port diameter resulted in higher flow rates (both in water and porcine vitreous), but only to a certain limit. When the port diameter became larger than the internal lumen of the vitrectomy cutter, less effect was noted on the overall flow rate . In the same study, when assessing different vitrectomy tips, a grater model demonstrated to be a safer option for vitreous shaving, avoiding direct cut of the retina. The authors stated that designing port geometries with the appropriate port size might allow a combination of maximal flow and accurate cutting.
The standard guillotine-shaped vitrectomy blade has been used for many years. However, its movement to cut the vitreous, with complete port closure, may result in flow instability, fluid acceleration and retinal traction [30, 32]. New blades have been designed in order to increase cut rate and overall surgery efficiency, while maintaining a safe environment.
Rizzo assessed a modified 23-gauge probe, with a “hole” in the standard guillotine blade, showing increased flow and cut rates, which could be associated with less retinal traction . However, using a similar blade design under egg albumen to simulate vitreous, Rossi et al. demonstrated higher particle acceleration than regular blades, which may lead to dangerous retinal movement .
Similarly, Claus Eckardt and Mitrofani Pavlidis, in conjunction with DORC, have developed a double-port two-dimensional cutter (TDC—Additional file 1: Video 1; available in 23, 25 and 27-gauge), which features a larger rectangular aperture in the inner lumen, with two sharp cutting edges, cutting vitreous in a forward and backward movement during each cycle, reaching rates of up to 16,000 cpm [14, 21, 27, 35]. No matter the position of the blade, the port is never occluded, leading to a duty cycle of almost 92%, with constant aspiration flow, even at higher cut rates . Osawa et al. carried out an experiment comparing the 27-gauge TDC and a standard 27-gauge probe, under BSS and porcine vitreous. The 27-gauge TDC flow rate under porcine vitreous was about 50% higher than the standard one. The BSS flow rate, remained constant regardless of the cut rate .
Cutter technology, duty cycle and their fluidics interaction
In general terms, in the core mode, when increasing the cut rate, the only time segment that can be reduced in order to achieve a higher cutting speed is the open port time, and the DC reduces. In the same manner, to increase the cut rate in the shave mode, the only segment that can be changed to obtain a higher cutting speed is the time the port remain closed. Consequently, the DC increases . The water flow rate tends to follow the DC pattern (i.e., if the DC decrease so does the water/BSS flow rate). The vitreous flow rate was reported with some different results in the literature. Diniz et al. have shown that under cutting speeds of up to 5000 cpm, the vitreous flow rate tends to increase even when the DC decreases. This could be explained by the vitreous fragmentation, resulting in less resistance to aspiration and improving the flow rate [24, 36, 38]. However, a new study from Abulon et al. assessing the vitreous behavior in porcine eyes during high-speed vitrectomy, with dual pneumatic cutters (23, 25 and 27-gauge), have demonstrated that vitreous flow rates at 7500-cpm, under biased open mode, was consistent with previously reported decrease in water and BSS flow rates with increasing cut rate. The authors stated that because resistance to flow is associated with increased vitreous viscosity, increased cut rate (which increases vitreous fragmentation and lowers vitreous resistance to flow) causes the fluid dynamics of vitreous to become similar to those of BSS (i.e., flow decreases with increased cut rate) although it still maintains efficient aspiration flow similar to 5000-cpm cutters. When using biased closed and 50/50 DC modes, the vitreous flow rate increased with higher cutting rates .
Despite the theoretical differences between the original spring return pneumatic cutters and the dual pneumatic driven cutters, Fernandes et al. recently compared the two mechanisms, under water and porcine vitreous, using two commercially available vitrectomy systems, in different probe sizes (20, 23, 25-gauge) and under different cutting speeds . The dual pneumatic cutter had a modulated DC set to the biased open mode. Interestingly, the authors were able to show similar DCs, vitreous and flow rates, with only small differences between the two systems, reinforcing the idea that both driven mechanisms may have similar performances under high cutting rates.
Small gauge overall considerations
Small gauge instruments (23, 25 and 27-gauge) are the procedure of choice of most of the vitreoretinal surgeons. Shorter operation time, less postoperative inflammatory reaction and conjunctival scarring, fast postoperative recovery, potential self-sealing wounds and less vitreoretinal traction are among the main advantages over 20-gauge vitrectomy. The initial hypotony concerns when using 23 and 25-gauge instruments following sutureless surgery were overcome with angled or two-step techniques of wound construction, although complete self-sealing sclerotomies are still not achievable in every single case [7–11]. Regarding the rate of endophthalmitis in sutureless microincision surgery (23 and 25-gauge), a systematic review by Govetto et al. did not find an increased risk when compared to 20-gauge surgery, although the authors recommended caution when interpreting the results due to the small number of events reported .
Major drawbacks of 25-gauge vitrectomy, when it was first introduced in 2002, were decreased illumination and instrument stiffness. New light sources managed to “put some light in the dark” (see next section) and probe modifications enhanced the instrument rigidity, although, by its own nature, it continues to be more fragile than 23-gauge. The smaller diameter, with decreased flow rates when compared to 20 and 23-gauge probes, actually contribute to the port based flow limiting and combined to high cutting rates and duty cycle control, results in less pulsatile vitreoretinal traction and enhance safety, as proposed by Steve Charles [20, 47]. An entire line of vitrectomy accessories is available and the indications cover the whole spectrum of vitreoretinal pathology .
Twenty-three gauge vitrectomy introduced by Eckardt, in 2005, came to address some of those early issues with 25-gauge probes, mainly concerning the instrument flexibility, lower flow rates and decreased illumination. The 23-gauge instruments had a smaller diameter than 20-gauge and were more rigid than 25-gauge, providing better illumination and facilitating the access to the peripheral vitreous (by eye rotation), without having to worry about instrument bending. Initially adopted for macular procedures, 23-gauge now incorporates the whole range of vitreoretinal procedures and is the device of choice by the majority of surgeons around the world according to 2014 ASRS pat survey.
Twenty-seven-gauge sclerotomies, with a smaller diameter (0.4 mm for 27-gauge; 0.5 mm for 25-gauge; 0.6 mm for 23-gauge) can be made perpendicular to the sclera and no angled or two-step techniques are required. The benefits of less inflammatory reaction, fast wound closure, less vitreous incarceration and fast postoperative recovery have potentially improved. The flow rates, as expected, have reduced when compared to 23 and 25-gauge systems, but increased cutting speed, combined with duty cycle control and newer cutter/blade shapes (e.g., two-dimensional cutter, by DORC International) as previously mentioned, have brought it to acceptable rates. Also, the rigidity was enhanced by the introduction of reinforcement sleeves (e.g., stiffness sleeve by Alcon, USA). Much like during the introduction of 23 and 25-gauge, 27-gauge instruments were initially used for selected cases. The indications, however, have already expanded to more complex surgeries. The smaller vitrectomy diameter and the port aperture closer to the tip can be positioned between narrow spaces, allowing membrane dissection and serving as a multifunctional instrument. The concerns about lack of endoillumination (see next section) have been resolved by the introduction of powerful Xenon, LED and future laser light sources [7, 8, 12, 13, 18, 49].
Numerous advances in endoillumination have occurred in the last decade, including the release of more powerful light sources, the usage of light filters to enhance tissue visualization and safety and the integration of chandeliers into complicated cases to allow bimanual surgery.
Light filters can be used to improve safety and possibly enhance tissue visualization. Various light sources over the last 10 years have incorporated filters that can be used to exponentially increase the safety calculations of the light source. The Synergetics Photon, DORC EVA and B&L Stellaris PC all have incorporated some variant of a yellow filter to allow this improved safety. The filters have also been felt to possibly enhance tissue visualization. A multicenter study using the 3 filters (Amber, Green, Yellow) on the B&L Stellaris PC was performed a few years ago within which the surgeons were asked to grade the quality of the view obtained with each of the filters in place during different stages of a vitrectomy procedure. Although the baseline color of the light source was universally accepted as good for all parts of surgery some of the interesting surgeons preferences were: the preference of the Amber filter for Air Fluid exchanges (it was felt to reduce glare) and also for peeling the internal limiting membrane (ILM) when Brilliant Blue dye was used and the preference of the Green filter to remove the ILM with most other dyes .
The use of vital dyes may also be related to phototoxicity and damage to the neuroretina and to the retinal pigment epithelium (RPE). These type of substances, such as indocyanine green (ICG), tryplan blue (TB), brilliant blue (BB) and other commercially available dyes, are used to enhance tissue visualization (e.g., internal limiting membrane, epiretinal membrane and vitreous) during vitreoretinal surgery, in what has been known as chromovitrectomy [51–53]. However, after tissue staining, they may interact with light sources and induce photosensitizing effects at the retinal surface by an overlap of the emission spectrum of the light source and the absorption band of the vital dye used during the vitrectomy procedure. An increased number of free radicals would be released and could lead to retinal and RPE damage [54, 55]. Haritoglou et al, staining retinas from postmortem human donor eyes with 0.5% ICG were able to demonstrate damage to the inner retinal layers after illumination using the halogen light source, probably due to the overlap of the light emitted from the light device (between 380 and 760 nm) and the light absorbing-properties of ICG (maximum absorption beyond 600 nm and fewer absorption at lower wavelengths of 500 nm) . Costa et al. also reported interesting findings when assessing the absorbance spectra of nine vital dyes (ICG, TB, BB, bromophenol blue, congo red, light green, fast green, indigo carmine and evans blue) diluted in three solvents (saline solution, glucose 5% and water) and their overlap with different light sources. In addition to the fact that the absorbance spectra varied with the solvent used, the authors have shown that the greatest overlap was found with integrated laser pathway (Photon Xenon; Synergetics Photon) and halogen lamp (Grieshaber GLS; GLS Corp.), and the least overlap was found with mercury vapor lamp (Photon 2; Synergetics). The lowest overlap values among the dyes were observed with ICG prepared in physiological saline solution, followed by indigo carmine, which showed low values for all three solvents compared with other dyes . The surgeon must be aware that regardless of the substance chosen, as mentioned by Farah et al. , intravitreal injection of a vital dye poses a dose-dependent toxicity to the retinal tissue and the interaction with a light source may contribute to exert further retina damage.
Another important concept to understand in endoillumination is the brightness, which involves calculation of the output of a light source, using a power meter. The results are then modulated to our photopic response curve, to actually obtain the brightness perceived by us (expressed in lumens). The older halogen/metal halide light sources, when using a 20 gauge light probe had a power output of around 8 lumens. With the release of 25-gauge vitrectomy, many surgeons immediately complained about the lack of light. Testing revealed that the original light sources only had an output of 2–4 lumens in 25-gauge which was the initial impetus to the development of the stronger Xenon and Mercury Vapor light sources . The stronger Xenon and Mercury Vapor light sources also allowed for clinically useful chandeliers and lighted instrumentation to be created, such as illuminated laser probes and vitrectomy picks, both contributing to a more assistant independent surgery, specially when working at the far periphery, where additional scleral depression by the assistant would usually be necessary.
Chandeliers, with multiple designs, have allowed “true” bimanual surgery and have a retinal threshold time in the order of hours, even when working at full output, which in summary, defines it’s significant safety. Additional file 4: Video 4 (courtesy of Dr. Oshima) shows an example of the 27-gauge Oshima Vivid Chandelier (Synergetics, USA). Although it has taken some time to integrate into clinical practice, recent data shows 75% of graduating retina fellows in North America will now regularly use a chandelier for complicated vitrectomies (unpublished data presented at the Retinal Fellows Forum, Chicago, 2014).
Recently, LED (light-emitting diode) light sources were introduced on the EVA platform (DORC, Netherlands) and the Versavit (Synergetics, USA). These LED bulbs offer the advantage of an extremely long life span of more than 10,000 h and allow surgeons the ability to titrate the color of their light source. There is also a laser light source which will be shortly released into the market (Katalyst Surgical, USA) incorporating 3 laser light sources which can be tuned to change the color of the light and provide a power capability on another level from the previous Xenon, Mercury Vapor and LED light sources. This increase in power will allow the usage of very small optical fibers, which can then be incorporated into our even smaller gauge instrumentation (25-gauge/27-gauge) to provide lighted 27-gauge instruments.
Wide-angle viewing systems
A clear and wide view is essential during vitreoretinal surgery. This was made possible with the wide-angle viewing systems (WAVs) initially introduced in the 1980s [56–58] and which are continuously under development. The WAVs allow a panoramic view of the retina based on the indirect ophthalmoscopic principle. A lens gives an inverted image, which is then reinverted by a prismatic device, generally connected to the microscope. Access to the peripheral vitreous is provided, even in the presence of small pupils and corneal opacity, improving both surgical efficiency and safety. There are two main types of WAVs: contact lens and non-contact lens [59–65].
Contact lens WAVs
Wide field contact lenses
Field of view (°)
Single use surgical WF
Non-contact lens WAVs
Non-contact wide field viewing systems
Approximate maximum field of view (°)
BIOM (HD disposable lens)
OFFSIS 120 D
Merlin wide angle lens
Resight 128 D lens
PWL 132 D lens
EIBOS 2 (132D) Moller-Wedel
The combination of a contact and a non-contact wide field system may also offer advantages. Chihara et al. designed a prototype contact lens, with zero power, used in combination with a non-contact wide angle system and showed it not only prevented the cornea from becoming dry, but led to a smooth corneal surface, providing a good quality of view . Some other studies assessed the simultaneous use of a magnifying contact lens and a non-contact WAVs: a wider field of view, along with no corneal dehydration and the potential ability to rapidly switch to a magnified macular view were observed [60, 61].
The non-contact WAVs have also been used along with scleral buckle procedures for the treatment of rhegmatogenous retinal detachment, under chandelier or slit lamp type endoillumination. Some possible advantages over the regular indirect ophthalmoscopy were mentioned: the image was not inverted, easier access to retinal breaks with dynamic scleral depression, even in small pupil eyes, and the ability of sharing the procedure image with medical staff and students. Pointed drawbacks were the risk of endophthalmitis, touching the lens with the illumination probe and vitreous wick from the scleral incision [62, 68, 69].
Vitreoretinal surgery is a constant changing field. The advances in cutter technology, endoillumination and WAVs over the years were noticeable and the efforts in the development of new instruments most lead to a better surgical performance while increasing safety. New studies on endoillumination are being conducted and will soon show the latest safety patterns of light sources from different commercially available devices. More studies comparing the newest surgical blades, 27-gauge probes and the regular 23-gauge and 25+ gauge systems are necessary to allow consistent conclusions. As technology improves, the next WAVs will certainly enhance our ability to access peripheral vitreoretinal pathology while providing high definition images during surgical procedures.
transconjunctival sutureless vitrectomy
vitreous infusion suction cutter
balanced salt solution
constant flow blade
internal limiting membrane
wide-angle viewing systems
Advanced Visual Instruments
Sensor Medical Technology
binocular indirect ophthalmoscopy microscopy
optic fiber free intravitreal surgery system
PRCO drafted, wrote the review, revised and submitted the manuscript; ARB advised the writing of the manuscript and revised; DRC advised, wrote the review and revised the manuscript. All authors read and approved the final manuscript.
Dr. David Chow has the following financial interests or relationships to disclose: consultant to Alcon, Bayer, DORC, Synergetics, and Katalyst; a lecturer for Allergan, DORC and Optovue.
This study did not receive any government and non-government support.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
- Machemer R. The development of pars plana vitrectomy: a personal account. Graefes Arch Clin Exp Ophthalmol. 1995;233(8):453–68.View ArticlePubMedGoogle Scholar
- Machemer R, Buettner H, Norton EW, Parel JM. Vitrectomy: a pars plana approach. Trans Am Acad Ophthalmol Otolaryngol. 1971;75(4):813–20.PubMedGoogle Scholar
- Charles S. The history of vitrectomy: innovation and evolution. Retina Today. 2008:27–9.
- Fujii GY, De Juan E Jr, Humayun MS, Chang TS, Pieramici DJ, Barnes A, et al. Initial experience using the transconjunctival sutureless vitrectomy system for vitreoretinal surgery. Ophthalmology. 2002;109(10):1814–20.View ArticlePubMedGoogle Scholar
- Fujii GY, De Juan E Jr, Humayun MS, Pieramici DJ, Chang TS, Awh C, et al. A new 25-gauge instrument system for transconjunctival sutureless vitrectomy surgery. Ophthalmology. 2002;109(10):1807–12 (discussion 13).View ArticlePubMedGoogle Scholar
- Eckardt C. Transconjunctival sutureless 23-gauge vitrectomy. Retina. 2005;25(2):208–11.View ArticlePubMedGoogle Scholar
- Osawa S, Oshima Y. 27-Gauge vitrectomy. Dev Ophthalmol. 2014;54:54–62. doi:10.1159/000360449.View ArticlePubMedGoogle Scholar
- Oshima Y, Wakabayashi T, Sato T, Ohji M, Tano Y. A 27-gauge instrument system for transconjunctival sutureless microincision vitrectomy surgery. Ophthalmology. 2010;117(1):93–102. doi:10.1016/j.ophtha.2009.06.043 (e2).View ArticlePubMedGoogle Scholar
- Mitsui K, Kogo J, Takeda H, Shiono A, Sasaki H, Munemasa Y, et al. Comparative study of 27-gauge vs 25-gauge vitrectomy for epiretinal membrane. Eye (Lond). 2016;30(4):538–44. doi:10.1038/eye.2015.275.View ArticleGoogle Scholar
- Recchia FM, Scott IU, Brown GC, Brown MM, Ho AC, Ip MS. Small-gauge pars plana vitrectomy: a report by the American Academy of Ophthalmology. Ophthalmology. 2010;117(9):1851–7. doi:10.1016/j.ophtha.2010.06.014.View ArticlePubMedGoogle Scholar
- Nagpal M, Paranjpe G, Jain P, Videkar R. Advances in small-gauge vitrectomy. Taiwan J Ophthalmol. 2012;2(1):6–12. doi:10.1016/j.tjo.2012.01.001.View ArticleGoogle Scholar
- Khan MA, Shahlaee A, Toussaint B, Hsu J, Sivalingam A, Dugel PU, et al. Outcomes of 27 gauge microincision vitrectomy surgery for posterior segment disease. Am J Ophthalmol. 2016;161:36–43. doi:10.1016/j.ajo.2015.09.024 (e1–e2).View ArticlePubMedGoogle Scholar
- Watanabe A, Tsuzuki A, Arai K, Gekka T, Tsuneoka H. Treatment of dropped nucleus with a 27-gauge twin duty cycle vitreous cutter. Case Rep Ophthalmol. 2016;7(1):44–8. doi:10.1159/000443721.View ArticlePubMedPubMed CentralGoogle Scholar
- Osawa S, Oshima Y. Innovations in 27-gauge vitrectomy for sutureless microincision vitrectomy surgery. Retina Today. 2014:42–5.
- Acar N, Kapran Z, Unver YB, Altan T, Ozdogan S. Early postoperative hypotony after 25-gauge sutureless vitrectomy with straight incisions. Retina. 2008;28(4):545–52. doi:10.1097/IAE.0b013e318162b008.View ArticlePubMedGoogle Scholar
- Woo SJ, Park KH, Hwang JM, Kim JH, Yu YS, Chung H. Risk factors associated with sclerotomy leakage and postoperative hypotony after 23-gauge transconjunctival sutureless vitrectomy. Retina. 2009;29(4):456–63. doi:10.1097/IAE.0b013e318195cb28.View ArticlePubMedGoogle Scholar
- Shimada H, Nakashizuka H, Hattori T, Mori R, Mizutani Y, Yuzawa M. Incidence of endophthalmitis after 20- and 25-gauge vitrectomy causes and prevention. Ophthalmology. 2008;115(12):2215–20. doi:10.1016/j.ophtha.2008.07.015.View ArticlePubMedGoogle Scholar
- Rizzo S, Barca F, Caporossi T, Mariotti C. Twenty-seven-gauge vitrectomy for various vitreoretinal diseases. Retina. 2015;35(6):1273–8. doi:10.1097/IAE.0000000000000545.View ArticlePubMedGoogle Scholar
- Steel DH, Charles S. Vitrectomy fluidics. Ophthalmologica. 2011;226(Suppl 1):27–35. doi:10.1159/000328207.View ArticlePubMedGoogle Scholar
- Charles S. Fluidics and cutter dynamics. Dev Ophthalmol. 2014;54:31–7. doi:10.1159/000360446.View ArticlePubMedGoogle Scholar
- Stalmans P. Enhancing visual acuity. Dev Ophthalmol. 2014;54:23–30. doi:10.1159/000360445.View ArticlePubMedGoogle Scholar
- Hassan TS. Advances in vitreous cutters. Retina Today. 2014:108–10.
- Rizzo S, Genovesi-Ebert F, Belting C. Comparative study between a standard 25-gauge vitrectomy system and a new ultrahigh-speed 25-gauge system with duty cycle control in the treatment of various vitreoretinal diseases. Retina. 2011;31(10):2007–13. doi:10.1097/IAE.0b013e318213623a.View ArticlePubMedGoogle Scholar
- Diniz B, Fernandes RB, Ribeiro RM, Lue JC, Teixeira AG, Magalhaes O, et al. Analysis of a 23-gauge ultra high-speed cutter with duty cycle control. Retina. 2013;33(5):933–8. doi:10.1097/IAE.0b013e3182733f38.View ArticlePubMedPubMed CentralGoogle Scholar
- Magalhaes O Jr, Chong L, DeBoer C, Bhadri P, Kerns R, Barnes A, et al. Vitreous dynamics: vitreous flow analysis in 20-, 23-, and 25-gauge cutters. Retina. 2008;28(2):236–41. doi:10.1097/IAE.0b013e318158e9e0.View ArticlePubMedGoogle Scholar
- Teixeira A, Chong LP, Matsuoka N, Arana L, Kerns R, Bhadri P, et al. Vitreoretinal traction created by conventional cutters during vitrectomy. Ophthalmology. 2010;117(7):1387–92. doi:10.1016/j.ophtha.2009.11.004 (e2).View ArticlePubMedGoogle Scholar
- Pavlidis M. Two-dimensional cutting (TDC) vitrectome: in vitro flow assessment and prospective clinical study evaluating core vitrectomy efficiency versus standard vitrectome. J Ophthalmol. 2016. doi:10.1155/2016/3849316.Google Scholar
- Pastor S, Bonshek R, Luciane I, Zambrano I, Carlin P, Stanga PE. New prototype of ultrasound harmonics vitrector (UHV) histophatological findings: first report. Investig Ophthalmol Vis Sci. 2015;56(7):394.Google Scholar
- Stanga PE, Pastor S, Zambrano I, Carlin P. New prototype of ultrasound harmonics vitrector (UHV) fluidics analysis: first report. Investig Ophthalmol Vis Sci. 2015;56(7):385.Google Scholar
- Rossi T, Querzoli G, Angelini G, Malvasi C, Iossa M, Placentino L, et al. Introducing new vitreous cutter blade shapes: a fluid dynamics study. Retina. 2014;34(9):1896–904. doi:10.1097/IAE.0000000000000143.View ArticlePubMedGoogle Scholar
- DeBoer C, Fang S, Lima LH, McCormick M, Bhadri P, Kerns R, et al. Port geometry and its influence on vitrectomy. Retina. 2008;28(8):1061–7. doi:10.1097/IAE.0b013e3181840b64.View ArticlePubMedGoogle Scholar
- Rossi T, Querzoli G, Malvasi C, Iossa M, Angelini G, Ripandelli G. A new vitreous cutter blade engineered for constant flow vitrectomy. Retina. 2014;34(7):1487–91. doi:10.1097/IAE.0000000000000251.View ArticlePubMedGoogle Scholar
- Rizzo S. Performance of a modified vitrectomy probe in small-gauge vitrectomy. Retina Today. 2011:40–2.
- Hayafuji M, Hanamura Y, Niimura S. “Surgical cutter” (Kabushiki Kaisha Topcon), US Patent 5106364; 1992.
- Ducournau D, Warren KA, Stalmans P, Pertile G, Eckardt C, McCannel C. EVA: a state-of-the-art surgical system for phaco-vitrectomy. Retina Today Suppl. 2013;1–16.
- Ribeiro RM, Teixeira AG, Diniz B, Fernandes RB, Zhong Y, Kerns R, et al. Performance analysis of ultrahigh-speed vitreous cutter system. Retina. 2013;33(5):928–32. doi:10.1097/IAE.0b013e31826f069e.View ArticlePubMedGoogle Scholar
- Abulon DJ. Vitreous flow rates through dual pneumatic cutters: effects of duty cycle and cut rate. Clin Ophthalmol. 2015;9:253–61. doi:10.2147/OPTH.S71387.View ArticlePubMedPubMed CentralGoogle Scholar
- Diniz B, Ribeiro RM, Fernandes RB, Lue JC, Teixeira AG, Maia M, et al. Fluidics in a dual pneumatic ultra high-speed vitreous cutter system. Ophthalmologica. 2013;229(1):15–20. doi:10.1159/000343073.View ArticlePubMedGoogle Scholar
- Abulon DJ, Buboltz DC. Performance comparison of high-speed dual-pneumatic vitrectomy cutters during simulated vitrectomy with balanced salt solution. Transl Vis Sci Technol. 2015;4(1):6. doi:10.1167/tvst.4.1.6.View ArticlePubMedPubMed CentralGoogle Scholar
- Witmer MT, Dugel PU. Machines and cutters: constellation®. Dev Ophthalmol. 2014;54:1–7. doi:10.1159/000360442.View ArticlePubMedGoogle Scholar
- Abulon DJ, Buboltz DC. Porcine vitreous flow behavior during high-speed vitrectomy up to 7500 cuts per minute. Transl Vis Sci Technol. 2016;5(1):7. doi:10.1167/tvst.5.1.7.View ArticlePubMedPubMed CentralGoogle Scholar
- Brant Fernandes RA, Diniz B, Falabella P, Ribeiro R, Teixeira AG, Magalhaes O Jr, et al. Fluidics comparison between dual pneumatic and spring return high-speed vitrectomy systems. Ophthalmic Surg Lasers Imaging Retina. 2015;46(1):68–72. doi:10.3928/23258160-20150101-11.View ArticlePubMedGoogle Scholar
- Gandorfer A, Ducournau D, Luff A, Prunte C, Chawla S, Jundt G. Fluidics in modern vitrectomy. Highlights from an expert roundtable meeting. Ophthalmol Times Eur. 2010:1–15.
- Salcedo-Villanueva G, Morales-Canton V. Experience with modern vitrectomy machines. Retina Today. 2012:45–6.
- Michalewska Z, Ducournau D, Adelman RA. How do vitrectomy parameters influence the results of rhegmatogenous retinal detachments repair? EVRS RD Study No. 3. Acta Ophthalmol. 2014;92(5):e416–7. doi:10.1111/aos.12318.View ArticlePubMedGoogle Scholar
- Govetto A, Virgili G, Menchini F, Lanzetta P, Menchini U. A systematic review of endophthalmitis after microincisional versus 20-gauge vitrectomy. Ophthalmology. 2013;120(11):2286–91. doi:10.1016/j.ophtha.2013.04.010.View ArticlePubMedGoogle Scholar
- Chow DR. The evolution of endoillumination. Dev Ophthalmol. 2014;54:77–86. doi:10.1159/000360452.View ArticlePubMedGoogle Scholar
- Mura M, Barca F. 25-Gauge vitrectomy. Dev Ophthalmol. 2014;54:45–53. doi:10.1159/000360448.View ArticlePubMedGoogle Scholar
- Schaal S, Ozkok A, Nesmith B. 27-Gauge vitrectomy surgery: smaller is better. Retina Today. 2014:24–32.
- Ham WT Jr, Mueller HA, Ruffolo JJ Jr, Guerry D 3rd, Guerry RK. Action spectrum for retinal injury from near-ultraviolet radiation in the aphakic monkey. Am J Ophthalmol. 1982;93(3):299–306.View ArticlePubMedGoogle Scholar
- Rodrigues EB, Meyer CH, Kroll P. Chromovitrectomy: a new field in vitreoretinal surgery. Graefes Arch Clin Exp Ophthalmol. 2005;243(4):291–3. doi:10.1007/s00417-004-0992-x.View ArticlePubMedGoogle Scholar
- Farah ME, Maia M, Penha FM, Rodrigues EB. The use of vital dyes during vitreoretinal surgery—chromovitrectomy. Dev Ophthalmol. 2016;55:365–75. doi:10.1159/000438963.View ArticlePubMedGoogle Scholar
- Al-Halafi AM. Chromovitrectomy: update. Saudi J Ophthalmol. 2013;27(4):271–6. doi:10.1016/j.sjopt.2013.10.004.View ArticlePubMedPubMed CentralGoogle Scholar
- Haritoglou C, Priglinger S, Gandorfer A, Welge-Lussen U, Kampik A. Histology of the vitreoretinal interface after indocyanine green staining of the ILM, with illumination using a halogen and xenon light source. Investig Ophthalmol Vis Sci. 2005;46(4):1468–72. doi:10.1167/iovs.04-0838.View ArticleGoogle Scholar
- Costa Ede P, Rodrigues EB, Farah ME, Dib E, Penha F, Magalhaes O Jr, et al. Vital dyes and light sources for chromovitrectomy: comparative assessment of osmolarity, pH, and spectrophotometry. Investig Ophthalmol Vis Sci. 2009;50(1):385–91. doi:10.1167/iovs.08-2285.View ArticleGoogle Scholar
- Oshima Y. Choices of wide-angle viewing systems for modern vitreoretinal surgery. Retina Today. 2012;37–42.
- Spitznas M. A binocular indirect ophthalmomicroscope (BIOM) for non-contact wide-angle vitreous surgery. Graefes Arch Clin Exp Ophthalmol. 1987;225(1):13–5.View ArticlePubMedGoogle Scholar
- Peyman GA. A new wide-angle irrigating contact lens for pars plana vitrectomy. Can J Ophthalmol. 1988;23(3):150.PubMedGoogle Scholar
- Chihara T, Kita M. New type of antidrying lens for vitreous surgery with a noncontact wide-angle viewing system. Clin Ophthalmol. 2013;7:353–5. doi:10.2147/OPTH.S42443.PubMedPubMed CentralGoogle Scholar
- Ohji M, Tada E, Futamura H. Combining a contact lens and wide-angle viewing system for a wider fundus view. Retina. 2011;31(9):1958–60. doi:10.1097/IAE.0b013e31822a900b.View ArticlePubMedGoogle Scholar
- Ohno H. Combined use of high-reflective index vitrectomy meniscus contact lens and a noncontact wide-angle viewing system in vitreous surgery. Clin Ophthalmol. 2011;5:1109–11. doi:10.2147/OPTH.S23621.View ArticlePubMedPubMed CentralGoogle Scholar
- Kita M, Fujii Y, Kawagoe N, Hama S. Scleral buckling with a noncontact wide-angle viewing system in the management of retinal detachment with undetected retinal break: a case report. Clin Ophthalmol. 2013;7:587–9. doi:10.2147/OPTH.S42923.View ArticlePubMedPubMed CentralGoogle Scholar
- Inoue M. Wide-angle viewing system. Dev Ophthalmol. 2014;54:87–91. doi:10.1159/000360453.View ArticlePubMedGoogle Scholar
- Chalam KV, Shah VA. Optics of wide-angle panoramic viewing system-assisted vitreous surgery. Surv Ophthalmol. 2004;49(4):437–45. doi:10.1016/j.survophthal.2004.04.010.View ArticlePubMedGoogle Scholar
- Park SW, Kwon HJ, Kim HY, Byon IS, Lee JE, Oum BS. Comparison of scleral buckling and vitrectomy using wide angle viewing system for rhegmatogenous retinal detachment in patients older than 35 years. BMC Ophthalmol. 2015;15:121. doi:10.1186/s12886-015-0109-9.View ArticlePubMedPubMed CentralGoogle Scholar
- Shah VA, Chalam KV. Self-stabilizing wide-angle contact lens for vitreous surgery. Retina. 2003;23(5):667–9.View ArticlePubMedGoogle Scholar
- Shah VA, Chalam KV. Autoclavable wide-angle contact lens for vitreous surgery. Am J Ophthalmol. 2004;137(2):359–60. doi:10.1016/s0002-9394(03)00870-5.View ArticlePubMedGoogle Scholar
- Aras C, Ucar D, Koytak A, Yetik H. Scleral buckling with a non-contact wide-angle viewing system. Ophthalmologica. 2012;227(2):107–10. doi:10.1159/000333101.View ArticlePubMedGoogle Scholar
- Imai H, Tagami M, Azumi A. Scleral buckling for primary rhegmatogenous retinal detachment using noncontact wide-angle viewing system with a cannula-based 25 G chandelier endoilluminator. Clin Ophthalmol. 2015;9:2103–7. doi:10.2147/OPTH.S91339.View ArticlePubMedPubMed CentralGoogle Scholar