- Letter to the Editor
- Open Access
Heads-up 3D vision system for retinal detachment surgery
International Journal of Retina and Vitreous volume 3, Article number: 46 (2017)
Here we present procedural and safety results from our initial experience using 3D heads-up vitrectomy for retinal detachment surgery. The comparison with our outcomes from previous standard surgery shows how the new 3D system seems safe and effective.
The application of a three-dimensional imaging system (3D) to retinal surgery is the latest and more exciting advance in the field. Previous studies demonstrated the technical feasibility of 3D surgery [1, 2]. This new imaging system offers many advantages including a more physiological “heads-up” position for the surgeon, a very high image definition even at wider magnifications, ad hoc digital filters (e.g., to enhance vitreous visibility), lower endoillumination levels , and the same view between the audience (e.g., nurses, fellows, students) and the surgeon [1, 2]. Following their release in the market, many surgeons started to operate routinely using a 3D system. Although the 3D system is different from traditional surgery, no direct comparison between these two methodologies is provided by the current literature.
The present study aimed to compare the outcomes of a 3D system with traditional ones in the setting of retinal detachment (RD) surgery.
From April 1st, 2017, all RD cases were routinely performed using the Ingenuity 3D system (Alcon, Fort Worth, Texas, USA). All cases were performed using a standard technique (25 gauge vitrectomy plus endolaser and gas/silicone oil tamponade) by a single experienced vitreoretinal surgeon (MC). An additional movie shows a case of rhegmatogenous RD operated using the 3D system (see Additional file 1). The surgical technique did not differ from previous surgery performed using a traditional system. We compared the 3D Group with a control group including all cases performed from January the 1st 2017 to March the 31st 2017 (Fig. 1).
We collected a total of 7 cases in the 3D group. Six (86%) of them were primary simple cases, while 1 (14%) of them was a complex recurrent one. Traditional group consisted of 15 cases of which 1 (7%) was a complicated recurrent case, and 14 (93%) were simple primary cases.
Immediate procedural success (defined as complete re-attachment at day 1 after surgery) was achieved in all cases for both groups. At 30 days after surgery, no cases of re-detachment were recorded in the 3D group, while one case occurred in the control group (p = 0.74). None of the eyes experienced any major post-operative complication. Five patients in the 3D group and two patients in the control one required medications for augmented intraocular pressure at day 1 (p = 0.56). The mean ± standard deviation of surgery time was 55 ± 35 min for the 3D group and 62 ± 28 min for the control group (p = 0.07). Mean endoillumination power during the procedure was 10 and 45% for groups 3D and control, respectively (p < 0.0001). In 9 eyes (60%) of the control group, diluted Triamcinolone was injected to improve the visualization of vitreous remnants. Conversely, vitreous staining was not performed in any eye of the 3D group (p = 0.01).
RD surgery is one of the most challenging procedures for the vitreo-retinal surgeon. Proper visualization during the surgery may influence the outcome dramatically. Some years ago, the introduction of non-contact wide field viewing systems led to considerable improvement in the field . The 3D systems provide a new viewing experience and offer several advantages compared to traditional systems (Table 1). This is, to the best of our knowledge, the first report to investigate the outcomes of RD repair using a 3D system. In our small initial series, we found no significant differences in procedural success rates compared to traditional surgery. Interestingly, the 3D system allowed avoiding triamcinolone vitreous staining. This could be related to the magnified high-resolution image, and to the use of digital filters (e.g., green filter) that enable the surgeon to visualize the vitreous remnants optimally. The avoidance of some surgical steps (i.e., vitreous staining) may explain the non-significant tendency towards shorter procedure time. However, we acknowledge that surgical time is influenced by the difficulty of each singular case. The homogeneity cannot be guaranteed by a non-randomized retrospective comparison and, therefore, differences in operating time between the two systems should be interpreted with caution.
The digital image enhancement enabled to reduce the endoillumination power significantly minimizing the phototoxicity to the retinal pigmented epithelium cells . All cases were performed using a light probe and, therefore, 3D visibility using other systems (i.e., chandelier light pipe, diaphanoscopic extraocular illumination) remains still indeterminate.
Due to the retrospective design, some variables (e.g., visualization of retinal holes and epiretinal membranes, visibility under air) were not recorded in our clinical charts and, therefore, we are not able to provide a direct comparison between the two imaging modalities for these points. We feel that 3D system, however, is not inferior to conventional surgery under these specific aspects. Due to the enhanced depth of field, multiple planes are contemporary in focus and this allows to have a wide field of view. Although the system has a latency 80 ms later than standard microscope, no significant delay was noted when performing intraocular procedures.
This report reflects the initial experience of a single center. Our findings need to be confirmed in larger multicenter, prospective studies. Nonetheless, in our experience, 3D visualization system seems to be as safe and effective as the traditional one providing all the advantages of digital over an analogic platform. In addition, it permits to avoid phototoxic risks related to endoillumination.
Eckardt C, Paulo EB. Heads-up surgery for vitreoretinal procedures: an experimental and clinical study. Retina. 2016;36:137–47. doi:10.1097/IAE.0000000000000689.
Dutra-Medeiros M, Nascimento J, Henriques J, Barrao S, Fernandes-Fonseca A, Aguiar-Silva N, Moura-Coelho N, Agoas V. Three-dimensional head-mounted display system for ophthalmic surgical procedures. Retina. 2017;37:1411–4. doi:10.1097/IAE.0000000000001514.
Adam MK, Thornton S, Regillo CD, Park C, Ho AC, Hsu J. Minimal endoillumination levels and display luminous emittance during three-dimensional heads-up vitreoretinal surgery. Retina. 2016. doi:10.1097/IAE.0000000000001420.
Landers MB, Peyman GA, Wessels IF, Whalen P, Morales V. A new, non-contact wide field viewing system for vitreous surgery. Am J Ophthalmol. 2003;136:199–201.
van den Biesen PR, Berenschot T, Verdaasdonk RM, van Weelden H, van Norren D. Endoillumination during vitrectomy and phototoxicity thresholds. Br J Ophthalmol. 2000;84:1372–5.
MC, CLS, FB, and AR participated in study design. AR and CLS carried out the literature search. MC, CLS, and AR drafted the manuscript. GQ and FB critically reviewed and revised the manuscript. All authors read and approved the final manuscript.
The authors declare that they have no competing interests.
Availability of data and materials
Material submitted for publication has not been previously reported and is not under consideration for publication elsewhere.
Consent for publication
All authors read and approved the final manuscript.
Ethics approval and consent to participate
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Coppola, M., La Spina, C., Rabiolo, A. et al. Heads-up 3D vision system for retinal detachment surgery. Int J Retin Vitr 3, 46 (2017) doi:10.1186/s40942-017-0099-2
- Heads-up surgery
- Retinal detachment