- Open Access
Endoscopic vitreoretinal surgery: principles, applications and new directions
International Journal of Retina and Vitreous volume 5, Article number: 15 (2019)
To analyze endoscopic vitreoretinal surgery principles, applications, challenges and potential technological advances.
Microendoscopic imaging permits vitreoretinal surgery for tissues that are not visible using operating microscopy ophthalmoscopy. Evolving instrumentation may overcome some limitations of current endoscopic technology.
Transfer of the fine detail in endoscopic vitreoretinal images to extraocular video cameras is constrained currently by the caliber limitations of intraocular probes in ophthalmic surgery. Gradient index and Hopkins rod lenses provide high resolution ophthalmoscopy but restrict surgical manipulation. Fiberoptic coherent image guides offer surgical maneuverability but reduce imaging resolution. Coaxial endoscopic illumination can highlight delicate vitreoretinal structures difficult to image in chandelier or endoilluminator diffuse, side-scattered lighting. Microendoscopy’s ultra-high magnification video monitor images can reveal microscopic tissue details blurred partly by ocular media aberrations in contemporary surgical microscope ophthalmoscopy, thereby providing a lower resolution, invasive alternative to confocal fundus imaging. Endoscopic surgery is particularly useful when ocular media opacities or small pupils restrict or prevent transpupillary ophthalmoscopy. It has a growing spectrum of surgical uses that include the management of proliferative vitreoretinopathy and epiretinal membranes as well as the implantation of posterior chamber intraocular lenses and electrode arrays for intraretinal stimulation in retinitis pigmentosa. Microendoscopy’s range of applications will continue to grow with technological developments that include video microchip sensors, stereoscopic visualization, chromovitrectomy, digital image enhancement and operating room heads-up displays.
Microendoscopy is a robust platform for vitreoretinal surgery. Continuing clinical and technological innovation will help integrate it into the modern ophthalmic operating room of interconnected surgical microscopy, microendoscopy, vitrectomy machine and heads-up display instrumentation.
Operating microscope ophthalmoscopy provides stereoscopic, high resolution, widefield extraocular imaging for vitreoretinal surgery. Microendoscopic ophthalmoscopy is a parallel technology that offers non-stereoscopic, video-monitor-based, ultra-high-magnification or panoramic intraocular imaging [1, 2]. Endoscopy permits surgery for vitreoretinal disorders that are difficult or impossible to observe with transpupillary imaging because of their anatomic location or ocular media problems [3,4,5]. It also permits continuation of operating microscope procedures that might otherwise be interrupted by problems precluding transpupillary ophthalmoscopy .
The visual experience of current intraocular endoscopy is quite different from that of surgical microscopy. Endoscopy bypasses ocular media optical aberrations and pathology. It can capture images in close proximity to tissue targets, allowing it to detect minute details that would require adaptive optics [7,8,9] augmentation to resolve with extraocular surgical microscope ophthalmoscopy. In essence, microendoscopy is an invasive alternative to adaptive optics imaging with performance limitations that include vitreous cavity clarity and intraocular probe caliber constraints that currently restrict the resolution of intraocular optical images transferred to extraocular digital monitors.
Safe and effective use of non-stereoscopic endoscopy requires training and new competences that include judging tissue distances on 2-dimensional (2-D) video monitor images with non-stereoscopic cues such as tissue target size and structure, motion parallax, and changes in tissue color or estimated texture. Surgical and technological innovation continue to drive the evolution of endoscopic vitreoretinal surgery decades after its clinical introduction. In this analysis of ophthalmic endoscopy, we (1) summarize its basic principles, (2) review current and developing clinical applications, (3) discuss surgical challenges and complications and (4) describe emerging technology.
Ophthalmic endoscopy principles
Current ophthalmic endoscopy surgical systems consist of a multifunction microendoscope and a wheeled cart housing a central console (base unit) and a high-resolution 2-D video monitor (Fig. 1) [2, 10]. The base unit incorporates an illumination source (usually xenon), a high-resolution video camera, interfaces for microendoscope fiberoptic cabling, and connections for video output to remote monitors. It can also contain equipment for digital image processing and a laser source for photocoagulation [1, 4, 5].
Fiberoptics: image guides, fiber bundles and laser fibers
A fiberoptic bundle consists of numerous optical fibers running parallel to each other. Each fiber has a central core surrounded by transparent cladding material that has a lower index of refraction. In widely used step-index fibers, total internal reflection limits light propagating down an optical fiber to its central core. Gradient index (GRIN) fibers constrain light to their central cores using an index of refraction that decreases radially outward from the fiber’s central axis to its periphery [11, 12].
Fiberoptic bundles are classified as coherent if fibers have the same position relative to each other at the beginning and end of a bundle. Coherent bundles can serve as image guides, preserving the information in transmitted optical images (Fig. 2) . Incoherent optical fiber bundles that don’t retain the relative positions of optical fibers are easier and less expensive to manufacture. They are useful for illumination but not transmitting optical images.
Laser photocoagulation radiation (typically 810 nm infrared radiation) is delivered by the central console’s laser source via a single, small diameter optical fiber . The laser fiber is enclosed in a protective sleeve before it combines with illumination and imaging optical bundles to form a cable extending to the tip of the microendoscope’s rigid handpiece-probe. Shorter wavelength green (532 nm) or yellow (577 nm) laser sources can be used to provide more ophthalmoscopically prominent lesions at lower tissue irradiances (power/area) [14, 15].
Microendoscope versus operating microscope imaging
Operating microscopes are used to view ophthalmoscopic images produced by contact or wide-angle non-contact lens systems [16,17,18]. These images are usually located in air between the lens system and the microscope. Increasing operating microscope magnification can improve the surgeon’s visualization of vitreoretinal detail in an ophthalmoscopic image but it cannot create optical information that isn’t already present in that image. Resolution in eyepiece or heads-up surgical microscope images is limited primarily by the optical performance of the ophthalmoscopic lens system and the patients’ ocular media, not by the operating microscope or video systems attached to it.
Endoscopy differs from operating microscopic ophthalmoscopy because magnifying an endoscopic image by moving the intraocular probe closer to a tissue target does create new optical information. The objective lens system mounted at the tip of the intraocular probe forms a high-resolution tissue image on the adjacent end-surface of the fiberoptic image guide (Fig. 2) . That image is the intraocular equivalent of the extraocular ophthalmoscopic image viewed in operating microscopy. Microendoscopy can permit inspection of (1) minute uveovitreoretinal details beyond the resolution of contemporary, extraocular non-adaptive-optics operating microscopy and (2) inspection of uveal and very anterior vitreoretinal regions that aren’t observable in transpupillary ophthalmoscopy even with scleral depression [3,4,5]. Table 1 compares ophthalmoscopy using microendoscopes and surgical microscopes.
Fiberoptic versus non-fiberoptic microendoscopes
In a contemporary ophthalmic fiberoptic microendoscope, optical fibers for illumination, imaging and optional laser photocoagulation extend from their console interfaces through a handpiece to the distal end of an intraocular probe (Fig. 2). The fiberoptic image guide relays the initially high-resolution optical image on its distal intraocular tip to its proximal end, which is interfaced with a digital video camera (Fig. 2). Resolution lost in this fiberoptic transmission depends largely on the number of optical fibers in the image guide (individual optical fibers correspond to screen pixels ).
Outside the endoscope’s rigid metallic handpiece-probe, fiberoptic bundles are encased in a mechanically-protective sleeve. Fibers in these cables can break if the cables are bent excessively or otherwise misused. Each broken optical fiber in the image guide eliminates a video pixel, producing a black spot on the monitor that reduces the endoscope’s resolution and its clinical lifetime.
Smaller caliber microendoscope probes are currently produced by reducing the number of image guide optical fibers. Reducing that number decreases a microendoscope’s (1) optical resolution because fewer pixels of ophthalmoscopic image information are transferred to the video camera and (2) field of view because the reduced diameter image guide captures a smaller area of the ophthalmoscopic image for a particular objective lens system (the intraocular end of the image guide bundle serves as an optical field stop).
Straight, curved and bent-tip rigid microendoscope probes have been developed . Probes are currently available in 19-, 20-, 23- and 25-gauge sizes [4, 5, 19]. Current smaller caliber microendoscopes have fewer image guide optical fibers, lower resolution and smaller imaging fields. Intraocular depth of field ranges from roughly 0.75–40 mm, permitting high magnification when the endoscope probe is adjacent to tissues and a panoramic intraocular view when it’s close to the sclerotomy site . Wider perspectives are especially helpful for instrument localization in bimanual procedures.
Hopkins rod-lenses and GRIN lenses currently retain more resolution than fiberoptic bundles in intraocular images relayed to extraocular cameras [2, 4]. Hopkins rod-lens endoscopes use small diameter cylindrical lenses with large length-to-diameter ratios to transfer the ophthalmoscopic image from the distal (intraocular) tip of the probe to the proximal end of the handpiece . GRIN lens endoscopes replace conventional objective and relay lenses with small diameter GRIN lenses. A conventional lens refracts light using the curved surfaces of homogeneous optical material that has a single index of refraction. GRIN lenses refract light using inhomogeneous optical material with refractive indices that vary radially, longitudinally or spherically [11, 12]. These lenses are often fabricated with flat ends to facilitate their use in multiple-lens systems and bonding to fiberoptic components.
Current ophthalmic GRIN endoscope probe-handpiece units are attached directly to a video camera to exploit their high resolution. This endoscope-camera configuration is less maneuverable in surgery than a fiberoptic microendoscope. Additionally, GRIN systems are relatively fragile and their non-autoclavable camera must be draped separately for surgery [2, 21]. In essence, current GRIN microendoscopes achieve higher resolution by sacrificing the surgical convenience and current lower cost of fiberoptic devices. Table 2 compares the features of fiberoptic and gradient index lens ophthalmic microendoscopes.
Endoilluminators and chandelier devices are used in vitreoretinal operating microscope procedures to illuminate surgical fields , eliminating the potentially dazzling ocular media light reflexes of extraocular light sources. Surgical microscopes use accessory non-contact or contact lens systems to form extraocular ophthalmoscopic images from tissues that are obliquely illuminated by these devices. Conversely, endoscopes illuminate tissue targets coaxially, using an objective lens system to form an intraocular ophthalmoscopic image with light scattered directly back at them from tissue targets. Tissues can be illuminated at different viewing angles in microendoscopy, optimizing visualization of subtle vitreous structures poorly identified in non-coaxial illumination or hidden by the instrument shadows or glare it causes [4, 10, 23].
Microendoscopic ophthalmoscopy can circumvent ocular media obscuration, identify tiny uveovitreoretinal abnormalities and image intraocular areas difficult or impossible to view with contemporary operating microscope ophthalmoscopy. Numerous procedures exploit these capabilities.
Posterior chamber intraocular lenses
Intraocular lens (IOL) implantation is an important part of some pars plana vitrectomy (PPV) procedures. Sutured, sulcus-fixated posterior chamber IOLs offer effective post-operative visual rehabilitation in adult and pediatric patients. Endoscopy can assure proper sulcus haptic placement and IOL centration (Fig. 3) but increases procedure duration .
Endoscopic evaluation is helpful before keratoprosthesis surgery if a patient’s visual potential or vitreoretinal condition is uncertain [25, 26]. Endoscopic cyclophotocoagulation can be performed after implantation . In eyes with an opaque cornea following ocular trauma, endoscopic surgery can allow earlier diagnosis and treatment of occult retinal tears and detachment, shorter operation times and fewer procedures than operating microscopy with a temporary keratoprosthesis [28, 29]. Earlier treatment is particularly helpful in pediatric patients where delays can cause severe proliferative vitreoretinopathy (PVR) and amblyopia .
The Argus II implant is approved by the U.S. Food and Drug Administration for use in patients with retinitis pigmentosa over 25 years of age whose prior useful vision has declined to minimal or no light perception [31, 32]. The implant’s 60 electrode array is inserted through a 5.2 mm sclerotomy into the vitreous cavity after vitrectomy. The array is centered over the macula and tacked to its inner retinal surface [31, 32]. A spectacle-mounted camera transmits video data to the electrodes to stimulate inner retinal cells still functioning after photoreceptor degeneration. Endoscopy and intraoperative OCT can reduce retinal damage and other intraoperative complications by helping guide sclerotomy as well as electrode array positioning and tacking [31, 33].
Subretinal hemorrhages in macular degeneration
Large subretinal hemorrhages from neovascular age-related macular degeneration cause profound, acute vision loss. Endoscopic surgery can be used to evacuate the subretinal blood and remove contributing fibrovascular retinal pigment epithelium without the need for large retinotomies. In a small case series, visual acuity improved in three of five patients and PVR did not occur within a 12-month follow-up period .
Intraocular foreign bodies
Ocular trauma can compromise media clarity thereby impairing surgical microscope ophthalmoscopy. Endoscopy bypasses this problem, permitting prompt identification and management of vitreoretinal problems and intraocular foreign bodies [29, 35,36,37]. Endoscopy can shorten trauma-to-surgery and surgical time. Its outcomes are similar to those of operating microscopy with a temporary keratoprosthesis but avoids some of the logistical issues associated with temporary implant use .
Internal limited membrane peeling
Endoscopy was used to repair a macular hole-related retinal detachment in a patient with severe keratoleukoma and pathologic myopia. Epiretinal membrane and indocyanine green-assisted internal limiting membrane peeling were monitored with a 23-gauge microendoscope. Silicone oil was removed 6 months post-operatively and the retina remained attached 12 months after surgery .
Scleral depression in operating microscope procedures permits ciliary body and peripheral retinal visualization. Alternatively, endoscopy doesn’t require scleral depression which hypothetically might contribute to postoperative inflammation and proliferation [5, 39]. Endoscopy without scleral depression identified retinal tears or holes in 19 of 20 consecutive patients whose breaks were undetectable preoperatively by indirect ophthalmoscopy with scleral depression . The retina was reattached in all 20 patients after a single operation.
Anterior PVR can cause ciliary body fibrosis and detachment leading to hypotony and phthisis (Fig. 4) [40, 41]. Endoscopy is a valuable tool for assessing ciliary body status and determining when to perform silicone oil refilling [40, 42]. Endoscopic vitrectomy to remove PVR from the ciliary body can be helpful in some cases of chronic hypotony, particularly in younger patients or those with fewer previous procedures (Fig. 5) .
In endoscopy-assisted anterior hyaloid membrane dissection, a 30-gauge needle can be placed in the canal of Petit and filtered air injected to dissect the anterior hyaloid membrane from the lens capsule (Fig. 6). The vitrector can then be used to remove the anterior hyaloid and vitreous [43, 44].
Operating microscope ophthalmoscopy is hampered by reflections from air-fluid interfaces. Endoscopic surgery for posterior PPV with a partial air-fluid exchange permits posterior membrane peeling under water while air in the anterior part of the vitreous cavity flattens the detached retina . This technique benefits from the increased maneuverability in patient head-positioning of endoscopy compared to surgical microscopy.
Surgical microscope visualization of posterior PVR is better under water that than air but endoscopy can provide improved viewing in air (“atmospheric endoscopic technique”) by avoiding anterior segment image degradation [46, 47]. Subretinal proliferation in severe posterior PVR can be removed without the need for retinotomies by inserting a microendoscope beneath detached retina .
Surgical challenges and complications
Heads-up display of stereoscopic operating microscope fields on a three-dimensional (3-D) monitor can reduce the cervical strain of eyepiece usage . Monitor-based endoscopic surgery has the same benefit but operating on the 2-D representations of 3-D structures increases surgical visual and cognitive workloads . New skills that must be mastered include remaining oriented in a 2-D intraocular field, adjusting illumination for probe-tissue distances, staying in the center of the endoscope’s image circle when manipulating tissue, and assessing distances from vital intraocular structures using non-stereoscopic cues such as motion parallax, estimated structure size or perceived tissue texture.
Inadequate endoscope illumination can limit target tissue image quality. Chandelier lighting systems can be helpful in such situations by diffusely illuminating a larger intraocular volume. Their disadvantages include instrument shadows and reflections that can impair tissue visualization . An additional endoilluminator can be useful if retinal phototoxicity risks and glare are minimized.
Tissue visualization can be compromised by vitreous hemorrhage, especially in cases of proliferative diabetic retinopathy or PVR. A fluid-air exchange or use of a liquid perfluorocarbon (e.g., perfluoro-n-octane) may help resolve imaging issues. Visualization can also be impaired by endoscope probe tip fogging after a fluid-air exchange. Defogging can be accomplished by repositioning the tip in fluid or retracting and cleaning it.
Some endoscopically visible tissues may be difficult to reach with a vitrectomy instrument because of cannula positioning, imaging limitations or a patient’s crystalline or intraocular lens. Exchanging the port locations of the microendoscope with the vitrectomy probe or infusion line can help solve access problems in these situations.
Endoscopic surgery complications that are infrequent for rhegmatogenous retinal detachment repair occur more commonly in advanced disorders . In one series of 12 patients treated for severe endophthalmitis, iatrogenic retinal tears occurred in 4 eyes, including a giant flap tear related to vitreous base clean-up in one eye .
Iatrogenic secondary cataract can be a distressing problem. Secondary cataract occurs in 6–20% of pediatric patients undergoing lens-sparing operating microscope vitreous surgery, mostly due to crystalline lens-retina approximation or instrument-crystalline lens touch [51, 52]. Endoscopy permits a unique view of instrumentation positioning relative to the crystalline lens. Maneuvering to avoid contact with the posterior lens surface can be challenging. In general, the risk of iatrogenic cataract can be minimized if the surgeon avoids passing instrumentation behind the posterior pole of the crystalline lens (Fig. 7) .
Endoscopic probe diameters are restricted by the caliber limitations of intraocular surgery. Some of the capabilities non-ophthalmic endoscopy may be achievable with emerging technology or the sequential use of specialized microendoscopes and intraocular probes. Innovation in microendoscopy faces the same conundrum in ophthalmology as it does in other surgical areas: demand requires availability which requires demand to justify development, manufacturing and distribution costs.
Stereoscopic endoscopy has been available but not uniformly accepted in disciplines including urology, gynecology and otolaryngology [49, 53, 54]. The potential advantages of 3-D vs. 2-D endoscopy include shorter learning curves for surgery, increased recognition of tissue texture and curvature, fewer surgical errors from visual misperception, decreased operating times, more precise depth judgment and decreased perceived workload [49, 54,55,56]. A prototype 18-gauge stereoscopic ophthalmic microendoscope was reported decades ago .
New technologies that potentially facilitate small caliber 3-D endoscopy include replacing optical fiber image guides with video microchips mounted on the distal end of endoscope probes (“chip-on-the-tip” sensor elements rather than optical cores correspond to video monitor pixels) . Dual-channel systems that operate similar to the human stereopsis and single channel devices that use a lens array overlying a single microchip (“insect eye” or “plenoptic” camera) are both used in non-ophthalmic stereoendoscopic surgery [59, 60]. Evolving single-channel stereoendoscopic technology includes rotating deflector and single-lens, dual-aperture devices [61, 62].
Stereoscopic endoscopy has technical challenges that include realistically representing depth information despite the huge disparity between a surgeon’s interocular distance and the effective separation distance of left and right image channels at a stereoendoscope’s tip (“gigantism”) [63,64,65]. Equipment-related and operator stereopsis-dependent side effects reported by some users of non-ophthalmic 3-D endoscopic systems include dizziness, headache and spatial disorientation [49, 63, 66, 67].
Reflectance and fluorescence
Standard microendoscopic light sources provide broad-spectrum white light. Illumination is reflected back into a coaxial imaging channel (fiberoptic bundle, grin lens or video microchip). Specular reflections from insubstantial vitreous structures can help identify tissues that are largely invisible in the side-scattered light of operating microscope intraocular illuminators [4, 10].
Microendoscope illumination and imaging channels can be color-filtered to permit endoscopic monochromatic ophthalmoscopy  or fluorescein angiography . Indocyanine green (ICG) angiography and autofluorescence imaging could be performed with appropriate filter sets. ICG, trypan blue and other vital dyes have been used to stain structures in vitreoretinal operating microscope procedures [70, 71]. They could also be used in endoscopic chromovitrectomy.
Digital image processing
Vitreoretinal endoscopic images can be computer-enhanced in real-time before they are displayed on a video monitor. Image size, contrast and color balance adjustment have been available . Additional computer techniques for real-time enhancement of surgical images include spatial frequency modulation (filtering) [72, 73], tissue topology and texture recognition , adaptive illumination adjustment  and image stabilization .
Robotic microscopic and endoscopic surgery
Robotic endoscopic surgery is a mature discipline in urology, gynecology and other non-ophthalmic medical specialties . Early efforts are underway to develop specialized robotic devices for ophthalmic surgery or use mature non-ophthalmic technology in ophthalmic applications [78, 79].
Optical fiber flexibility and prisms
Ophthalmic microendoscopes currently use rigid straight or curved intraocular probes for vitreoretinal or more anterior surgery, respectively . Flexible, sterilizable endoscopes are used in other surgical disciplines where larger diameter endoscopes can accommodate channels for illumination, imaging, instrumentation and tip-bending control wires [80, 81]. Single-use illumination and laser probes are available for vitrectomy using fiberoptic bundles that curve when extended from a rigid intraocular housing .
Small diameter coherent image guides have been developed with enough flexibility to bend around a small coin without fiber breakage. The two ends of fiberoptic bundles are fused so that individual optical fibers maintain the same relative position at both ends of the bundle, thereby providing the coherence needed for image transmission. Outer cladding material elsewhere is removed (acid “leached”) to separate fibers and produce a highly flexible fiber bundle .
Improved flexibility in a coherent imaging bundle can increase microendoscope longevity. Flexible-tip, single-use fiberoptic image guides are currently too expensive for surgical use. Flexible-tip, multi-use multifunction probes with instrumentation channels and guide wires aren’t feasible for ophthalmic endoscopy because of intraocular probe caliber limitations. An alternative way to extend microendoscopic field of view is to incorporate a prism into the objective lens system at the distal end of a rigid, straight intraocular probe. This technology has been developed for GRIN microendoscope probes .
Large screen, ultra-high-definition 3-D monitors can be used as a heads-up display for surgical microscopes [48, 84, 85]. Ample screen space is available for additional intraoperative data including (1) real-time optical coherence tomography (OCT) or microendoscope imaging and (2) technical details such as intraocular and infusion pressures [48, 85]. Supplementary data can be available as needed but otherwise not distracting. Protocols driven by physician, government and/or industrial groups may eventually compel manufacturers to make data streams from different operating room imaging sources compatible as for example was done for medical images with the Digital Imaging and Communications in Medicine (DICOM) standard.
Microendoscopy is currently used primarily as an independent surgical platform. Periodic switching between viewing a small endoscopy monitor and operating microscope oculars is helpful for staying oriented in the surgical field and positioning endoscopic instrumentation. Large, multiple or split-screen, ultra-high-definition monitors can eliminate this task by displaying both information streams simultaneously and also improving endoscope display quality .
Operating microscope-integrated OCT (MIOCT) systems can currently display 2-D or 3-D OCT data in microscope oculars [86,87,88]. Supplemental intraoperative OCT data could be made available during microendoscope procedures from sources including: (1) MIOCT systems, (2) independent intraocular OCT probes or (3) very small diameter OCT probes integrated into microendoscopes. Intraocular OCT probes are in the developmental stage .
Microendoscopy permits effective surgery on intraocular structures impossible to view with transpupillary ophthalmoscopy because of their location or ocular media opacification. It’s range of vitreoretinal applications continues to expand. Developing technology promises further expansion of those capabilities as well as more effective integration of ophthalmic operating room microscope, microendoscope, vitrectomy machine and heads-up display instrumentation.
pars plana vitrectomy
optical coherence tomography
Digital Imaging and Communications in Medicine
surgical microscope integrated OCT
Uram M. Ophthalmic laser microendoscope endophotocoagulation. Ophthalmology. 1992;99:1829–32.
Uram M. Endoscopic surgery in ophthalmology. Philadelphia: Lippincott Williams & Wilkins; 2003.
Marra KV, Yonekawa Y, Papakostas TD, Arroyo JG. Indications and techniques of endoscope assisted vitrectomy. J Ophthalmic Vis Res. 2013;8:282–90.
Wong SC, Lee TC, Heier JS, Ho AC. Endoscopic vitrectomy. Curr Opin Ophthalmol. 2014;25:195–206.
Kawashima S, Kawashima M, Tsubota K. Endoscopy-guided vitreoretinal surgery. Expert Rev Med Dev. 2014;11:163–8.
Fisher YL, Heringer GC. Endoscopy for Vitreoretinal Surgery. In: Peyman GA, Meffert SA, Conway MD, Chou F, editors. Vitreoretinal surgical techniques. Malden: Blackwell Science Inc; 2001. p. 107–12.
Booth MJ. Adaptive optical microscopy: the ongoing quest a perfect image. Light Sci App. 2014;3:1–7.
Roorda A, Duncan JL. Adaptive optics ophthalmoscopy. Annu Rev Vis Sci. 2015;1:19–50.
Burns SA, Elsner AE, Sapoznik KA, Warner RL, Gast TJ. Adaptive optics imaging of the human retina. Prog Retin Eye Res. 2019;68:1–30.
Yeo DCM, Nagiel A, Yang U, Lee TC, Wong SC. Endoscopy for pediatric retinal disease. Asia Pac J Ophthalmol (Phila). 2018;7:200–7.
Meyer-Arendt JR. Introduction to classical and modern optics. Englewood Cliffs: Prentice Hall Inc; 1995.
Hecht E. Optics. San Francisco: Addison Wesley Inc; 2002.
Ford HD, Tatam RP. Characterization of optical fiber imaging bundles for swept-source optical coherence tomography. Appl Opt. 2011;50:627–40.
Mainster MA. Wavelength selection in macular photocoagulation. Tissue optics, thermal effects, and laser systems. Ophthalmology. 1986;93:952–8.
Mainster MA. Decreasing retinal photocoagulation damage: principles and techniques. Semin Ophthalmol. 1999;14:200–9.
Ho PC, Mainster MA, Dieckert JP, Tolentino FI. Fundus contact lenses for closed pars plana vitrectomy. Ophthalmology. 1983;90:106–14.
Mainster MA, Crossman JL, Erickson PJ, Heacock GL. Retinal laser lenses: magnification, spot size, and field of view. Br J Ophthalmol. 1990;74:177–9.
Chalam KV, Shah VA. Optics of wide-angle panoramic viewing system-assisted vitreous surgery. Surv Ophthalmol. 2004;49:437–45.
Raizada S, Al Kandari J, Al SK. Ophthalmic endoscope: a novel disgnostic imaging tool. Int J Diag Imag. 2017;4:9–15.
Cockett WS, Cockett AT. The Hopkins rod-lens system and the Storz cold light illumination system. Urology. 1998;51:1–2.
Arroyo JG. The role of endoscopy in vitreoretinal surgery today. Retina Today. 2013; 54–56.
Oshima YU. Chandelier endoillumination in vitreoretinal surgery. Retina Today. 2013;8:68–72.
Witmer MT, Chan RVP. Chandelier lighting during vitreoretinal surgery. Retina Today. 2012;7:35–7.
Olsen TW, Pribila JT. Pars plana vitrectomy with endoscope-guided sutured posterior chamber intraocular lens implantation in children and adults. Am J Ophthalmol. 2011;151(287–296):e282.
Farias CC, Ozturk HE, Albini TA, Berrocal AM, Amescua G, Betancurt C, et al. Use of intraocular video endoscopic examination in the preoperative evaluation of keratoprosthesis surgery to assess visual potential. Am J Ophthalmol. 2014;158(80–86):e82.
Fukuda M, Liu C. The role of intraocular video endoscopic fundal examination before keratoprosthesis surgery. Am J Ophthalmol. 2014;158:3–4.
Poon LY, Chodosh J, Vavvas DG, Dohlman CH, Chen TC. Endoscopic cyclophotocoagulation for the treatment of glaucoma in boston keratoprosthesis type II patient. J Glaucoma. 2017;26:e146–9.
Chun DW, Colyer MH, Wroblewski KJ. Visual and anatomic outcomes of vitrectomy with temporary keratoprosthesis or endoscopy in ocular trauma with opaque cornea. Ophthalmic Surg Lasers Imaging. 2012;43:302–10.
Ayyildiz O, Hakan DA. Comparison of endoscopic-assisted and temporary keratoprosthesis-assisted vitrectomy in combat ocular trauma: experience at a tertiary eye center in Turkey. J Int Med Res. 2018;46:2708–16.
Sabti KA, Raizada S, Kandari JA, Wani V, Gayed I, Kumar N. Applications of endoscopy in vitreoretinal surgery. Retina. 2008;28:159–66.
Finn AP, Grewal DS, Vajzovic L. Argus II retinal prosthesis system: a review of patient selection criteria, surgical considerations, and post-operative outcomes. Clin Ophthalmol. 2018;12:1089–97.
Hahn P, Fine HF. Practical concepts with the Argus II retinal prosthesis. Ophthalmic Surg Lasers Imaging Retina. 2018;49:742–6.
Ozmert E, Demirel S. Endoscope-assisted and controlled Argus II epiretinal prosthesis implantation in late-stage retinitis pigmentosa: a report of 2 cases. Case Rep Ophthalmol. 2016;7:315–24.
Kaga T, Kojima T, Yokoyama S, Sato H, Yoshida N, Ichikawa K. Subretinal endoscopic surgery to treat large subretinal hemorrhages secondary to age-related macular degeneration. Retina. 2018. https://doi.org/10.1097/IAE.0000000000002121.
Kawashima M, Kawashima S, Dogru M, Inoue M, Shimazaki J. Endoscopy-guided vitreoretinal surgery following penetrating corneal injury: a case report. Clin Ophthalmol. 2010;4:895–8.
Sabti KA, Raizada S. Endoscope-assisted pars plana vitrectomy in severe ocular trauma. Br J Ophthalmol. 2012;96:1399–403.
Colyer M. Endoscopy: an overlooked tool in the management of ocular trauma. Retina Today. 2016; 80–2.
Chen Y, Shen L, Zhao S, Wang L, Xu C. Internal limiting membrane peeling by 23-gauge endoscopy for macular hole retinal detachment in a pathological myopic eye. Ophthalmic Surg Lasers Imaging Retina. 2017;48:179–82.
Kita M, Yoshimura N. Endoscope-assisted vitrectomy in the management of pseudophakic and aphakic retinal detachments with undetected retinal breaks. Retina. 2011;31:1347–51.
Faude F, Wiedemann P. Vitreoretinal endoscope for the assessment of the peripheral retina and the ciliary body after large retinectomies in severe anterior PVR. Int Ophthalmol. 2004;25:53–6.
Lee GD, Goldberg RA, Heier JS. Endoscopy-assisted vitrectomy and membrane dissection of anterior proliferative vitreoretinopathy for chronic hypotony after previous retinal detachment repair. Retina. 2016;36:1058–63.
Boscher C, Kuhn F. An endoscopic overview of the anterior vitreous base in retinal detachment and anterior proliferative vitreoretinopathy. Acta Ophthalmol. 2014;92:e298–304.
Yu YZ, Zou YP, Zou XL. Endoscopy-assisted vitrectomy in the anterior vitreous. Int J Ophthalmol. 2018;11:506–11.
Kam YW, Funk RO, Barnard L, Ajlan RS. New endoscopic surgical approach for anterior hyaloid dissection in phakic and pseudophakic patients. Retina. 2019. https://doi.org/10.1021/acsinfecdis.8b00279.
Hattori T, Sonoda KH, Kinoshita S. Two useful techniques of pars plana vitrectomy using endoscope. Eye (Lond). 2006;20:1466–8.
Enaida H, Ikeda Y, Yoshida S, Nakao S, Hachisuka Y, Fujita K, et al. Clinical evaluation and feasibility of changing intraoperative visibility with a novel viewing filter system for human eye. Retina. 2013;33:1923–30.
Kaga T, Yokoyama S, Kojima T, Mitamura H, Mori T, Matsuda T, et al. Novel endoscope-assisted vitreous surgery combined with atmospheric endoscopic technique and/or subretinal endoscopic technique for rhegmatogenous retinal detachment with grade C proliferative vitreoretinopathy. Retina. 2018;. https://doi.org/10.1097/IAE.0000000000002121.
Yonekawa Y. Seeing the world through 3-D glasses. Retina Today. 2016; 54–60.
Sorensen SM, Savran MM, Konge L, Bjerrum F. Three-dimensional versus two-dimensional vision in laparoscopy: a systematic review. Surg Endosc. 2016;30:11–23.
Boscher CR. Endoscopy-assisted vitrectomy for severe endophthalmitis. Retinal Phys. 2012;9:38–58.
Maguire AM, Trese MT. Lens-sparing vitreoretinal surgery in infants. Arch Ophthalmol. 1992;110:284–6.
Choi J, Kim JH, Kim SJ, Yu YS. Long-term results of lens-sparing vitrectomy for stages 4B and 5 retinopathy of prematurity. Korean J Ophthalmol. 2011;25:305–10.
Schwab K, Smith R, Brown V, Whyte M, Jourdan I. Evolution of stereoscopic imaging in surgery and recent advances. World J Gastrointest Endosc. 2017;9:368–77.
Bickerton R, Nassimizadeh AK, Ahmed S. Three-dimensional endoscopy: the future of nasoendoscopic training. Laryngoscope. 2019. https://doi.org/10.1002/lary.27812.
Marcus HJ, Hughes-Hallett A, Cundy TP, Di Marco A, Pratt P, Nandi D, et al. Comparative effectiveness of 3-dimensional versus 2-dimensional and high-definition vs standard-definition neuroendoscopy: a preclinical randomized crossover study. Neurosurgery. 2014;74:375–80; discussion 380–371.
Sakata S, Grove PM, Hill A, Watson MO, Stevenson ARL. Impact of simulated three-dimensional perception on precision of depth judgements, technical performance and perceived workload in laparoscopy. Br J Surg. 2017;104:1097–106.
Eguchi S, Kohzuka T, Araie M. Stereoscopic ophthalmic microendoscope system. Arch Ophthalmol. 1997;115:1336–8.
Charalampaki P, Igressa A, Mahvash M, Pechlivanis I, Schick B. Optimal invasive key-hole neurosurgery with a miniaturized 3D chip on the tip: microendoscopic device. Asian J Neurosurg. 2013;8:125–31.
Tabaee A, Anand VK, Fraser JF, Brown SM, Singh A, Schwartz TH. Three-dimensional endoscopic pituitary surgery. Neurosurgery. 2009;64:288–93; discussion 294-285.
Singh A, Saraiya R. Three-dimensional endoscopy in sinus surgery. Curr Opin Otolaryngol Head Neck Surg. 2013;21:3–10.
Radfar E, Jang WH, Freidoony L, Park J, Kwon K, Jung B. Single-channel stereoscopic video imaging modality based on transparent rotating deflector. Opt Express. 2015;23:27661–71.
Bae SY, Korniski RJ, Shearn M, Manohara HM, Shahinian H. 4-mm-diameter three-dimensional imaging endoscope with steerable camera for minimally invasive surgery (3-D-MARVEL). Neurophotonics. 2017;4:011008.
Sakata S, Grove PM, Stevenson AR. Effect of 3-dimensional vision on surgeons using the da vinci robot for laparoscopy: more than meets the eye. JAMA Surg. 2016;151:793–4.
Kunert W, Auer T, Storz P, Braun M, Kirschniak A, Falch C. How much stereoscopic effect does laparoscopy need? Controlled, prospective randomized trial on surgical task efficiency in standardized phantom tasks. Surg Innov. 2018;25:515–24.
Mendiburu B. 3D movie making. Oxford: Focal Press; 2009.
Hofmeister J, Frank TG, Cuschieri A, Wade NJ. Perceptual aspects of two-dimensional and stereoscopic display techniques in endoscopic surgery: review and current problems. Semin Laparosc Surg. 2001;8:12–24.
Biddle M, Hamid S, Ali N. An evaluation of stereoacuity (3D vision) in practising surgeons across a range of surgical specialities. Surgeon. 2014;12:7–10.
Delori FC, Gragoudas ES, Francisco R, Pruett RC. Monochromatic ophthalmoscopy and fundus photography. The normal fundus. Arch Ophthalmol. 1977;95:861–8.
Uram M. Endoscopic fluorescein angiography. Ophthalmic Surg Lasers. 1996;27:849–55.
Costa Ede P, Rodrigues EB, Farah ME, Dib E, Penha F, Magalhaes O Jr, et al. Vital dyes and light sources for chromovitrectomy: comparative assessment of osmolarity, pH, and spectrophotometry. Invest Ophthalmol Vis Sci. 2009;50:385–91.
Bracha P, Ciulla TA, Baumal CR. Vital dyes in vitreomacular surgery. Ophthalmic Surg Lasers Imaging Retina. 2018;49:788–98.
Mainster MA. Contemporary optics and ocular pathology. Surv Ophthalmol. 1978;23:135–42.
Gonzalez RC, Woods RE, Eddins SL. Digital image processing using MATLAB. Knoxville: Gatesmark, LLC; 2009.
Lin J, Clancy NT, Elson DS. An endoscopic structured light system using multispectral detection. Int J Comput Assist Radiol Surg. 2015;10:1941–50.
Xia W, Chen ECS, Peters T. Endoscopic image enhancement with noise suppression. Healthcare Technol Lett. 2018;5:154–7.
Schoob A, Laves MH, Kahrs LA, Ortmaier T. Soft tissue motion tracking with application to tablet-based incision planning in laser surgery. Int J Comput Assist Radiol Surg. 2016;11:2325–37.
Gosrisirikul C, Don Chang K, Raheem AA, Rha KH. New era of robotic surgical systems. Asian J Endosc Surg. 2018;11:291–9.
de Smet MD, Naus GJL, Faridpooya K, Mura M. Robotic-assisted surgery in ophthalmology. Curr Opin Ophthalmol. 2018;29:248–53.
Edwards TL, Xue K, Meenink HCM, Beelen MJ, Naus GJL, Simunovic MP, et al. First-in-human study of the safety and viability of intraocular robotic surgery. Nat Biomed Eng. 2018;2:649–56.
Reavis KM, Melvin WS. Advanced endoscopic technologies. Surg Endosc. 2008;22:1533–46.
Kovaleva J, Peters FT, van der Mei HC, Degener JE. Transmission of infection by flexible gastrointestinal endoscopy and bronchoscopy. Clin Microbiol Rev. 2013;26:231–54.
Mansour SE. Retinal probes: refining our laser delivery tools. New Retina MD. 2015; 7.
Kim JK, Lee WM, Kim P, Choi M, Jung K, Kim S, et al. Fabrication and operation of GRIN probes for in vivo fluorescence cellular imaging of internal organs in small animals. Nat Protoc. 2012;7:1456–69.
Miyake K, Ota I, Miyake S, Tanioka K, Kubota M, Mochizuki R. Application of a newly developed, highly sensitive camera and a 3-dimensional high-definition television system in experimental ophthalmic surgeries. Arch Ophthalmol. 1999;117:1623–9.
Rezende F, Vila N. Unlocking the potential of 3-D surgery. Retina Specialist. 2018;4:25–31.
Carrasco-Zevallos OM, Keller B, Viehland C, Shen L, Waterman G, Todorich B, et al. Live volumetric (4D) visualization and guidance of in vivo human ophthalmic surgery with intraoperative optical coherence tomography. Sci Rep. 2016;6:31689.
Ehlers JP, Modi YS, Pecen PE, Goshe J, Dupps WJ, Rachitskaya A, et al. The DISCOVER study 3-year results: feasibility and usefulness of microscope-integrated intraoperative OCT during ophthalmic surgery. Ophthalmology. 2018;125:1014–27.
Kolb JP, Draxinger W, Klee J, Pfeiffer T, Eibl M, Klein T, et al. Live video rate volumetric OCT imaging of the retina with multi-MHz A-scan rates. PLoS ONE. 2019;14:e0213144.
Asami T, Terasaki H, Ito Y, Sugita T, Kaneko H, Nishiyama J, et al. Development of a fiber-optic optical coherence tomography probe for intraocular use. Invest Ophthalmol Vis Sci. 2016;57:568–74.
Martin A. Mainster, PhD, MD, FRCOphth is a consultant for Ocular Instruments, Inc. which manufactures ophthalmoscopic lenses. The other authors declare that they have no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Ajlan, R.S., Desai, A.A. & Mainster, M.A. Endoscopic vitreoretinal surgery: principles, applications and new directions. Int J Retin Vitr 5, 15 (2019). https://doi.org/10.1186/s40942-019-0165-z
- Operating microscope
- Laser photocoagulation
- Heads-up display
- Gradient index lens
- Hopkins rod-lens
- Coherent fiberoptic
- Motion parallax
- Intraocular lens
- Proliferative vitreoretinopathy
- Optical coherence tomography
- Surgical microscope integrated OCT
- Adaptive optics